dust test chamber sample pretreatment If the technical conditions of the sample are required, sample pretreatment should be carried out as required to keep the samples, Part of the dust test case protective test, usually with the anti-test test, after the sample waterproof test, must wait for it to dry after it can be placed, if the sample is not pretreated, then press GB2423. The provisions of 37 were placed at 2 h under normal atmospheric conditions:
B relative temperature: 45% -75%;
C Atmospheric pressure: 8 kPa-106 kPa.
2, sample enters
dust test chamber
Mounting of different protective grade samples:
A protective level is 1P5X, and the housing of the first type (Class A) sample in the case in the case of the surrounding atmospheric pressure is normal, and should be closely replaced. The position is provided with a pumping hole, and the gas on the sample is attached to the vacuum system with a hose.
C protect the level of 1 p6X, whether it is a Class A sample or Class B sample housing, To take a vacuum.4
The amount of dry talite powder calculated by each gram-meter dust test equipment (including the volume of the pipe and the working space) is placed on the screen.
5, the glass door of the test chamber
Checking the dust test equipment should be well sealed, and it should be able to prevent dust from extrusion.
For samples that need to be vacuum, when booting, while opening the blower and vacuum pumping system, adjusting the valve, making vacuum and air flow consistent with product technical conditions for technical conditions for unwanted empty samples Only the blower is only drifting, does not open the vacuum system.
7, replacement of talc
If the sample technical conditions require intermediate testing, after the test reaches the specified time, the blower of the vacuum system will stop the vacuum system, after the dust is completely settled, open the glass door of the dust box, do not take the test Intermediate detection is carried out in cases. For samples without intermediate testing, when the test time or pumping amount reaches the requirements, stop the action of the vacuum system and the blower, after the dust is completely settled, after removing the sample, in this section, paragraph 1 ldquo; pretreatment of the sample rdquo; under conditions (ie ldquo; normal atmospheric conditions) place 1H-2H.
9, post detection
A Check the position of the sample housing into the amount of dust and the position of dust settling.
B is required to inspect items such as electrical, mechanical properties according to the sample techniques.
10, supplementary description
A If the sample technical conditions are required to perform ldquo; initial measurement rdquo;
C According to the technical conditions of the sample, dust removal can be carried out with the same small sample as the large sample structure. Test, the test results represent the large sample.
D, according to the test, the test must be loaded when the test is required, and the dust test should be added during the dust test. The load on the sample.
E as ldquo; dust rdquo; before investing in the test equipment, it should be sieved according to the specified 200 mesh screen. Net screening and maintain a dry state.
g shouldDust test requirements, regular replacement of talcum powder.
H test sample suspension conditions before the test.CAUTION Vibrations may occur when booting, preventing the samples from falling and solid vibration from the hook due to suspension, thereby causing damage to the sample and damaged the dust box.
The linker of the Vacuum system must not be leaked, the gas flowmeter and the pressure gauge should be installed and used in the instrument,To check if the vacuum oil in the vacuum pump is insufficient, the air blower is suitable.
You can Send Message or email email@example.com to us, we will reply tu you within 24 hours.Now tell us your need,there will be more favorable prices!
No. 2578 Gu Dai Road, Shanghai