The testing machine for all QUV and QSQSQ-lab is equipped with a blackboard temperature sensor, monitoring and accurate holding of temperature in the test. For some models of QSQSQ-lab, you can replace two temperature sensors or simultaneously with the temperature sensors in the box.
The tested machine's blackboard temperature sensor needs to be calibrated regularly. The calibration of the QUV test machine simply uses any reference thermometer. The Q-SUN test box must be calibrated using a special blackboard thermometer. Calibration Thermometer itself needs to be calibrated, once a year.All QUV and QSQSQ-lab testing machines are equipped with Solar Eye solar eye radial illumination control system (except for QUV // basic testing machines). Solar Eye sun eye system can monitor and accurately maintain the test
The level of irradiation in the test.The sensor in the Solar Eye sun eye system must be regularly calibrated by the user. This work can be done with special calibration irradiance. The irradiance meter itself needs to be calibrated, once a year. QUV thermometer calibration
QUV blackboard temperature sensor needs to be calibrated by the user to ensure accurate, consistent. The calibration of the QUV blackboard temperature sensor is simple and can be calibrated with any standard reference thermometer.
V-2156-X QUV blackboard temperature sensor (replacement on-demand)
Q-sun thermometer calibration
* All QSQSQ-lab Blackboard Temperature Sensors need to be calibrated by the user to ensure accurate, consistent.* Calibrate the Blackboard Temperature Sensor is simple and only take a few minutes.
* Our calibration laboratory has passed the US Laboratory Certification Association (A2LA) or UKAS ISO 17025 certification in the US and UK.* Calibration Traceable to National Standards and Technology Research Institutes (NIST).
The aging light failure tester has an optional indoor air temperature sensor (including XE-1 type for the chiller).Calibrating the XE-1 indoor air temperature sensor is simple and can be calibrated with any standard reference thermometer.The sunlight aging tester and XE-3 model are deeply designed and improved due to low cost and relative humidity / room air temperature sensors. It should be replaced once a year without re-correction.
Common QSQSQ-lab thermometer zero components are as follows:
X-10235-X Q-SUN XE-1 / XE-3 black panel temperature sensor (on-demand replacement) ]
X-7778-X QSQSQ-lab Xe-1 / Xe-3 Insulated Blackboard Temperature Sensor (Demand Replacement)
XR-11103-X QSQSQ-lab B02 Blackboard Temperature Sensor (Date-Demand
XR-11109-X QSQSQ-lab B02 insulated blackboard temperature sensor (on-demand replacement)
X-10415-K QSQSQ-lab B02 / XE-3 relative humidity / indoor air sensor (Replace each year)
CT-202 / IBP QSQSQ-lab Insulation Black Calibration Temperature Meter and Case
CT -202 / recal thermometer recalibration service (recommended once a year)