qinsun technician gives you the following solutions:
First of all glass nozzles: Now the nozzles of salt fog test box are very small inner diameters made by quartz glass. I can't see it. We should check the glass nozzle.
1. Be careful with the nozzle and blow it off with the mouth. If it is not connected, it means that the nozzle is blocked. We blow out the impurities inside with the trachea against the mouth. If you can't blow it, use fine feathers.
2. If the nozzle is sprayed with water instead of fog, the diameter of the nozzle should be broken (the caliber becomes larger) so it is not fog.
3. In addition, you need to check the two tubes on the nozzle. If there is no wrong position, it is the nozzle of the salt fog test box. You must change the nozzle.
If you can't solve it, please ask the technical staff of the instrument manufacturer for professional inspection. Kunshan qinsun instrument specializes in the production of environmental testing equipment such as salt fog test boxes. Lifetime maintenance and consultation!