<?xml:namespace prefix = o /> [ 123]
[ 123]1 . Working environment:
A. Temperature 15
15° C 35
° C. B. Relative humidity: no greater than 85% . C.
Volve voltage, frequency 50Hz plusmn 1 Hz . D.
E. It is not suitable for open or sunshine. 2 . After the box was not damaged: A. Four foot should be flat and keep the horizontal.
The device and the spacing of the surrounding object or wall: before ≥ 900mm , left and right and top, ≥ 500mm .
Check if the power outlet meets the requirements (the power plug of the voltage device) is again inserted into the power plug. d.
d.On the right side of the drug stability test box (using the stent pad high), place the fitting in the upper side of the overflowing in the back of the back, drainage.
3 . Place the water tank (spare parts) on the right side of the box, add a water plastic tube to connect the water tank, and the other is slightly inserted into the right side of the box on the right side of the box. Here, the quick-connect connection is used. When removing, the blue wafer at the plus water should be tightened to the casing, and the water pipe can be opened.
a. To ensure the sensitivity of the water cup float inside, be sure to add pure water or distilled water.
5 . Put the water pump power supply into the pumping pump power interface socket 6.
General limited temperature setting is a high temperature limit by control panel Decision dial implementation, generally larger than the tank temperature value
° C ], Limited temperature setting ZUI is ° C + ° C
Beautiful side .
, Open , to turn off according to actual needs .
.can give a change in intensity . ( placing the test sample in the center of the boxWhen two tube light intensity