1. Check whether the gas source is normal.Second, check the pressure gauge, see if the pressure gauge is adjusted to the range, and if there is no need to adjust to the range within the range.
Third, check whether the spray switch hits i, if not hit, open the spray switch.Fourth, check whether the nozzle saline pipe and the trachea are reversed, if the reverse is adjusted back.
5, check if the tip is installed too low. If the tower is installed too low, adjust to a suitable position
6. Check if the salt tank is free, if so, add salt water .
Seven, check if the nozzle is blocked or ruptured. If it is blocked, please clean the nozzle and then put it up. If there is a rupture, please replace the nozzle, check whether the hose is broken.8. Check the suction salt tube
whether to drain saline, if salt water, please replace the salt water pipe.Nine, check whether the intake pipe is blocked or detached.
Ten, check if the solenoid valve is damaged, detecting: Whether the solenoid valve coil is measured by the multimeter.