Analysis of the high and low temperature test chamber with heating tube and humidification tube drying
high and low temperature test chamber When humidifying the drying alarm, it is possible to be caused by the following reasons: 1
, the water level of the machine humidified water disc will not be humidified The tube is completely immersed (the normal standard is about 5mm of water immersion), it is possible that the machine is placed enough, resulting in partially exposed water, need to adjust the foot foot of the machine, allow the machine to measure the machine completely
The float switch of the water level is damaged, and the water level is still closed, resulting in dry and humidity,
Keep checking if the pressure balance is blocked, there is a water in the water pipe, blocking the smooth balance of the balance 4 , super temperature The protector itself is damaged
High and low temperature test chamber
[ When the heating tube is dried, it is possible to be caused by the following reasons:
Open the control box side cover, see if the set value of the gas switch is correct, normal than the ZUI high temperature of the machine is 50 ~ 80 degrees [[ 3
If the heating tube and the humidification tube are done, the fault phenomenon is eliminated according to the above. Please timely with the Beijing QINSUNCircle After-sales service department, our company will resolve you * time.
You can Send Message or email email@example.com to us, we will reply tu you within 24 hours.Now tell us your need,there will be more favorable prices!
No. 2578 Gu Dai Road, Shanghai